Four transfer standards were used, two from NPL and two from PTB. Each of these two laboratories provided a noise standard on waveguide IEC R140 and a noise standard on coaxial PC-7 with a precision waveguide adapter. The calibration frequencies were 12.4 - 13.5 - 15.0 - 17.5 and 18 GHz. The quantity to measure was the excess noise ratio (ENR).
Final Report 2005-05-31
The project has been finished and results are available at the KCDB.
Progress Report 2000-12-30
The project was extended into a CCEM.RF key comparison designed as CCEM.RF-K9. Two national laboratories were added to the comparison loop, NIST and VNIIFTRI.
Measurements were started in April 1997 and were completed in December 2000.
The Draft A report (confidential report) combining both EUROMET 329 and CCEM.RF-K9 results, and issued by BNM-LNE/LAMA (ex BNM-LCIE), has been completed and submitted to the participants in January 2002.
The comparison is now under responsibility of the RF working group of the CCEM, to which the Draft B report will be directly submitted. The EUROMET project is finished.