We will compare bell and piston provers for the following flow rates:
0,25 m3/h, 0,5 m3/h, 1 m3/h, 2 m3/h, 4 m3/h, 6 m3/h, 8 m3/h, 10 m3/h, 12 m3/h, 15 m3/h, 18 m3/h, 20 m3/h and 25 m3/h using a rotary piston gas meter (G16 DN50, Triple from Precision Gas Measurement, NL) as a transfer standard.
The laboratories should carry out the calibration using their common procedures.
Each laboratory will be responsible for the transport of the transfer standard to the next laboratory. No other costs are expected.
- Test gas: air at atmospheric pressure
- Test temperature: (19.5 ÷ 22.5) °C
Progress Report 2022-05-24
The measurements were completed and all participants have submitted their results.
The coordination of the project was taken over by Claudia Berkmann (BEV, Austria) from Anna Wonaschütz (BEV, Austria).